Humans perceive color and appearance subjectively and differently, even with a trained eye. In a model for human color perception, the observer is represented as the human eye that receives light reflected or transmitted from an object and processed in the brain to become a color perception. This application note is a review of the calculation of CIE X, Y, Z color values while “standardizing” the human observer as a numerical representation of what the “average person” sees.
Defining the Standard Observer
In the 1920s two researchers from the UK, David Wright and John Guild, performed color matching experiments using human volunteers to develop and quantify the color ability of a standard, or average, human observer. In 1931 the CIE published the 2° Standard Observer based on their research. This Standard Observer is called 2° because, during the color matching experiment, subjects looked through a hole that allowed them a 2° field of view. At the time it was believed that all the color-sensing cones of the eye were located within a 2° arc of the fovea, located directly back from the retina of the eye. The experiments were performed by projecting colors from across the visible spectrum onto a screen. Multiple people matched each spectral color light using a combination of red, green, and blue lights. The curves generated from this data resulted in bar x, bar y and bar z functions, referred to as the 1932 CIE 2° degree Standard Observer.
By the 1960s, it was realized that cones in the human visual system covered a larger field of view than previously believed.The visual matching experiments were repeated by W. S. Stiles, J. M. Burch and N. I. Speranskaya using this wider field of view and in 1964, the CIE 10° Standard Observer was published. The 10° Standard Observer is recommended as best representing the spectral response of human observers. As shown in Figure 2, the 1931 2° Standard Observer is similar to the 1964 10° Standard Observer and was not discontinued with the 1964 release due to its incorporation in many industry product specifications. Today, the 1964 10° Standard Observer is recommended by the CIE as appropriate for most industrial color applications but the 1931 2° remains available as a measurement choice. The relative sizes of the two fields of view are shown in Figure 1.
The 1931 2° and 1964 10° Standard Observer functions are used in the calculation of color values to reflect the human response to the visible spectrum.
Defining Color Values
The CIE X, Y, Z tristimulus values are calculated from the CIE Standard Observer functions, a selected CIE illuminant and the reflectance or transmittance of the sample. At each wavelength, bar x, bar y and bar z functions are multiplied by the CIE illuminant. Then that value is multiplied by the reflectance or transmittance of the sample at each wavelength. The values for all the wavelengths are then summed and integrated to display CIE X, Y, Z color values.
CIE X, Y, Z tristimulus values are calculated as follows:
X = ∫(R or T) * illuminant factor *bar x of standard observer
Y = ∫(R or T) * illuminant factor * bar y of standard observer
Z = ∫(R or T) * illuminant factor * bar z of standard observer
where R = Reflectance or T=Transmittance for the spectral range of the instrument
Note that X, Y, and Z include factors for the CIE standard observer in their formulas. All other tristimulus color scales, such as Hunter L, a, b or CIE L*a*b*, are calculated from the CIE X, Y, Z color scale.
This note was provided as background information to relate human color vision to instrumental color measurements. Using instruments to measure color perception creates an objective and precise method to quantify a perceived by humans.
(See attached for the complete article)